Most COVID-19 vaccination programmes use LDV syringes to reduce vaccine wastage, but people are misreading them.
Find out why LDV syringes are being used, and how to read them properly!
LDV Syringe : What Is It?
Low dead volume (LDV) or low dead space (LDS) syringes and needles are specially designed to reduce wastage.
Most COVID-19 vaccination programmes are using these LDV / LDS syringes and needles to minimise vaccine wastage in each shot.
A normal syringe, for example, will have about 0.1 ml of fluid left after a full injection. A LDV syringe / needle can reduce that to just 0.003 ml!
These special syringes and needles allow a skilled healthcare worker to extract an extra dose or two from each vial of COVID-19 vaccine.
LDV Syringe : How To Read The Volume Correctly?
There have been a few cases where vaccine recipients have been given less than the recommended dose of a COVID-19 vaccine.
In other cases, it is likely that the vaccine recipients misread the volume of LDV syringes, which are different from regular syringes.
To avoid such misunderstandings, healthcare workers are now asked to show the vaccine recipient the pre-filled syringes before vaccination, and the empty syringe after vaccination.
However, vaccine recipients must first understand how much COVID-19 vaccine should be in the syringe, and how to read the volume correctly!
How Much Volume For Each Vaccine Shot?
Different vaccines have different recommended volumes. Most vaccines are titrated with a recommended dose volume of 0.5 ml, but not Pfizer.
|Pfizer COMIRNATY||0.3 ml|
|AstraZeneca Vaxzevria||0.5 ml|
|Sinovac CoronaVac||0.5 ml|
How To Read LDV Syringes
Low dead volume needles are used with regular syringes, and are read normally. So they are not a problem.
LDV syringes though, have a special long rubber plunger – the black part. Most people misread it by looking at the tip of this long plunger.
It should be read from the first ring of the LDV plunger, as the picture below shows.
If you are not aware of this, you would probably think the LDV syringe contained only 0.35 ml of vaccine. In reality, it contains 0.5 ml of the vaccine.
Don’t forget to also check the vial, to verify you are getting the right COVID-19 vaccine – to avoid potential mix-ups in vaccination centres that offer different COVID-19 vaccines.
Help Support My Work!
Name : Adrian Wong
Credit Card / Paypal : https://paypal.me/techarp
Bank Transfer : CIMB 7064555917 (Swift Code : CIBBMYKL)
Thank you in advanced!
- Failed To Get AstraZeneca 2.0? Check MySejahtera Anyway!
- AstraZeneca 2.0 Appointments Delayed But Will Come!
- Scam Alert : Herbal Visual Tea For Glaucoma / Cataract!
- Fact Check : Did Lim Kok Wing Die From COVID-19?
- Fact Check : Our Immunity Drops After COVID-19 Vaccination?
- Can You Get The COVID-19 Vaccine With Other Vaccinations?
- Vaccine Underdose : What Should You Watch Out For?
- AstraZeneca Vaccine : Can You Ask For Earlier Second Dose?
- AstraZeneca Vaccine Blood Clots : What To Look For?
- Fact Check : US Withdraws After Chinese Jamming Attacks?
- Fake COVID-19 Vaccine Injection In Malaysia Debunked!
- Fact Check : New Deadly African COVID-19 Variant?
- Is COVID-19 Spreading Through Gas / Petrol Pumps?
- Fake News Alert : Sinovac Vaccines For Selangor Residents!
- Ivermectin Can Prevent / Cure COVID-19? Not So Fast!
- Can You Take Painkillers Before / After COVID-19 Vaccination?
- Dr Clo Disinfectant Sticks : Do They Work Against COVID-19?
- Why You Should NOT Reject The Vaccine You’re Offered!